Download the OneTouch Reveal® mobile app for FREE!
The Bluetooth® word mark and logos are registered trademarks owned by Bluetooth SIG, Inc. and any use of such marks by LifeScan Scotland Ltd. and its affiliates is under license. Other trademarks and trade names are those of their respective owners.
Apple, the Apple logo, iPhone, iPad and iPod touch are trademarks of Apple Inc., registered in the US and other countries. App Store is a service mark of Apple Inc.
Google Play and the Google Play logo are trademarks of Google LLC.
Before using the OneTouch Verio Flex® meter to test your blood glucose, carefully read the Owner’s Booklet and inserts that come with the components of the system.
How do I turn on the OneTouch Verio Flex® meter?
Press and hold ‘OK’ until the start-up test screen appears. Once the device is on, release ‘OK’. You can also turn the meter on by inserting a OneTouch Verio® test strip into the test strip port.
If you see any missing segments within the start-up screen, there may be a problem with the meter. Contact Customer Service at 1-800-663-5521
What batteries do the OneTouch Verio Flex® meter use?
Your OneTouch Verio Flex® meter uses one CR2032 lithium coin cell battery.
What are the low and high range limits and what do they mean?
Your meter uses low and high range limits to tell you when your result is below, above or within the limits that you set in the meter.‡ The meter comes with pre-set range limits that can be changed as recommended by your healthcare team. The pre-set low range limit is 4mmol/L and the pre-set high range limit is 10mmol/L.
‡The low and high range limits you set apply to all glucose test results. This includes tests taken before or after mealtimes, medications and around any other activities that may affect your blood glucose.
How can I change the low and high range limits?
With the meter on, press and hold ∧ and ∨ at the same time. The SET screen will appear with the current low range limit displayed, and the number and Range Indicator arrow will flash. The low and high range limits can be adjusted but be sure to talk to your healthcare professional about the range limits that are right for you.
If you change your range limits, this will only affect whether future test results are displayed as below, above or within your range limits. Changing your range limits does not affect how past results are displayed.
How can I use Bluetooth® on my meter?
You can connect your meter with a compatible wireless device running the OneTouch Reveal® mobile app. To turn Bluetooth® on§, press ‘OK’ and ∧ at the same time. After you open the OneTouch Reveal® mobile app on your mobile device, follow the on-screen instructions and enter the PIN shown on your meter screen when prompted.
§The Bluetooth® word mark and logos are registered trademarks owned by Bluetooth SIG, Inc., and any use of such marks by LifeScan Canada Ltd. is under license. Other trademarks and trade names are those of their respective owners. For further details please refer to the Owner’s Booklet.
Can I transfer my readings to a computer?
Your meter can work with the OneTouch Reveal® web app which provides simple visuals to help you see your blood sugar readings along with important things like food and insulin. You can track meals, medications and activities in a personalized logbook.
Hardware and software requirements needed to use the OneTouch Reveal® web app can be found by clicking "Learn More".
*Of those who expressed a preference and had experience using the 30G OneTouch Delica lancet and at least one other lancing device, 72% of 1739 respondents agreed. Based on an online survey conducted in USA in 2016.